The TS4085-I is an indoor temperature sensor. It can monitor transmission site equipment temperature and can also be used for better HVAC performance and efficiency.
Measurement accuracy is ±1.8ºF (±1ºC) for temperatures between ‑40ºF and +185ºF (‑40ºC and +85ºC). The output voltage can be easily converted to either ºF or ºC using the following formulas:
T (ºF) = [(V0 – 1.25) x 180] + 32
T (ºC) = (V0 – 1.25) x 100
The sensor’s operating voltage range is 6 to 15 VDC. It features low self-heating, linear +18 mV/ºF (+10 mV/ºC) scale factor, less than 100 µA drain current at 15 VDC, and low impedance output.
Use the following polynomial equation linearization coefficients in your Davicom for this temperature sensors:
For F degrees: A = 0, B = 180, C = -193
For C degrees C: A = 0, B = 100, C = -125